• Join over 1.2 million students every month
• Accelerate your learning by 29%
• Unlimited access from just £6.99 per month
Page
1. 1
1
2. 2
2
3. 3
3
4. 4
4

# Callibrate a Hall Probe: Planning

Extracts from this document...

Introduction

Planning Exercise

Design a laboratory experiment to calibrate an uncalibrated Hall probe and use it to investigate how the magnetic flux density mid-way between opposite poles of two permanent bar magnets varies with the separation of the bar magnets.

Calibrating the Hall probe

To calibrate the Hall probe, Helmholtz coils should be used as a uniform field is produced.

Helmholtz coils[1]

The magnitude of the flux density[2] in the region of uniform field is given by

where B = flux density (T)

µ0 = permeability of free space (4π x 10-7 Hm-1)

n = no. of turns on coil

I = current in each coil (A)

R = radius of each coil (m)

Formula 1

The magnitude of the Hall voltage[3] measured by a Hall probe is given by

where VH = Hall voltage (V)

I = current in Hall probe (I)

B = magnetic flux density (T)

n = no. of electrons per unit volume (e/m3)

e = electron charge (1.6 x 10-19 C)

d = thickness of film in Hall probe (m)

Formula 2

Middle

Position the Hall probe in the centre so that it is perpendicular to the direction of the field.

1. Record the voltage reading from the milli-voltmeter.
2. Use Formula 1 to find a value for the corresponding magnetic flux density.
3. Increase the current going through he Helmholtz coil to 1.0A.
4. Measure the voltage reading and use Formula 1.
5. Repeat this for coil currents of 1.5A, 2.0A and 2.5A.
6. Record values for voltage and magnetic flux density using a data logger and create an automated graph.
7. Use this graph to find the corresponding magnetic flux density for any value of voltage shown by the milli-voltmeter.

NB

• The current through the coils must not exceed 3A to avoid overheating.
• The current through the Hall probe must not exceed 0.4A.
• The Hall probe should be connected to a DC power supply so that a constant current is provided and therefore a constant temperature. This increases the reliability of the results as changes in temperature can affect the measurements.
• A milli-voltmeter increases the accuracy of the voltage readings.
• Run the positive and negative leads of the coils together so that the fields they are generating are cancelled out.
• A data logger ensures accurate results.

Conclusion

Read the voltage and use the graph created during calibration to work out the magnetic flux density.Increase the distance between the magnets to 0.02m.Read the voltage and use the graph to find the magnetic flux density.Take readings for distances of 0.01m, 0.02m, 0.03m, 0.04m, 0.05m and 0.06m.Record the results of magnetic flux density and create a graph of distance between magnets against magnetic flux density to find a relationship, which should follow the inverse square law.

NB

• Ensure that this experiment is performed away from other magnetic materials so that results are not affected.

References

• http://hyperphysics.phy-astr.gsu.edu/hbase/magnetic/helmholtz.html
• http://physicslabs.phys.cwru.edu/EM/Manual/pdf_version/MAG.pdf
• A-Level Physics 4th Edition by Roger Muncaster – Chapter 41, Pg 615
• A-Level Physics 4th Edition by Roger Muncaster – Chapter 41, Pg 637

[1] http://hyperphysics.phy-astr.gsu.edu/hbase/magnetic/helmholtz.html

[2] A-Level Physics 4th Edition by Roger Muncaster – Chapter 41, Pg 615

[3] A-Level Physics 4th Edition by Roger Muncaster – Chapter 41, Pg 637

[4] http://physicslabs.phys.cwru.edu/EM/Manual/pdf_version/MAG.pdf

This student written piece of work is one of many that can be found in our AS and A Level Fields & Forces section.

## Found what you're looking for?

• Start learning 29% faster today
• 150,000+ documents available
• Just £6.99 a month

Not the one? Search for your essay title...
• Join over 1.2 million students every month
• Accelerate your learning by 29%
• Unlimited access from just £6.99 per month

# Related AS and A Level Fields & Forces essays

1. ## Industrial Exercise

Regenerative Braking: Regenerative breaking is used on all trams as it is a great way to convert what would be wasted energy back into electricity and back into the system. It is especially useful in Sheffield supertrams as there are many trams on the track throughout the day and the

2. ## Investigating a factor affecting the voltage output of a transformer.

This is because as the primary voltage is increased, so the alternating current in the primary coil also increases. This creates an alternating magnetic field in the primary coil, resulting in the magnetic field lines changing direction more frequently. They are cut more times in a given time period by

1. ## To calibrate an uncalibrated Hall probe and use it to investigate how the magnetic ...

This force can be calculated using the equation: Fmagnetic = B Q v where B is the magnetic flux density, Q is the charge of the electron and v is the speed at which the electrons move Once the electrons move and create a potential difference across the opposite sides

2. ## Investigation of determination of flux density of magnets(plane)

More generally, magnetic flux is defined by a scalar product of the magnetic field and area element vector. Magnetic flux density used to describe the amount of flux flowing through unit area at a point in the field. The symbol used for flux density is B, and its unit is the tesla, T.

1. ## To see how the number of coils on an electromagnet affect its strength.

I expected my results to come out correct because I followed the procedures of the experiment correctly. However, I noticed that there were some anomalies in my final out of results. As you can see in the graph, when I picked up paper clips on the 35th turn of coil, 30 paper clips were picked up.

2. ## I determined the band gap of an undoped sample of germanium. I Measuring the ...

"The Hall potential difference V is associated with the electric field across the strip width d"(1) The potential difference is given by If a voltmeter is connected across the width of the strip, the potential difference can be measured. The voltmeter can also indicate which side of the strip has

1. ## The history of Magnets and electromagnets.

deposits of magnetic ore (magnetite) have been discovered at various places one of these being Labrador. Pieces of this material are called natural magnets because they exhibit the properties of magnetism without any special treatment. The earth is considered to be a huge natural magnet because it possesses the same properties as smaller magnets.

2. ## Maglev Trains And The Technology Behind Them (magnetism)

By using them we can find the size of the magnet that is needed to levitate two people at 20mm. First to find the force we multiply the weight of two people at 70N each by the gravitational constant 9.8 and then multiply this figure by a safety factor of two we get 2744N.

• Over 160,000 pieces
of student written work
• Annotated by
experienced teachers
• Ideas and feedback to